Maintenance advisory: CLF generator scheduled for replacement around Mar. 15. Auger solar panel used
to power local trigger electronics may need to be moved or disconnected.
Dec. 15 2016
Fixed a bug in local trigger program that caused the system to
skip sending a T3 for a coincident event. The program is throttled
to allow one local event per 5 mins (due to limited baud rate of
Auger South). This results in about 300-400 local events per day.
Local trigger issues: as seen from the diagnostic plots here
the local trigger system (TICTAC) is not stable. The rate increases by roughly
10x during the day time. This suggests a light leak or unstable threshold level
in the comparators. Further investigation required.